About the name of the axis
The horizontal direction is the X axis, the Y axis, and the vertical direction is the Z axis, and the θ axis (θZ) is rotated around the Z axis;
Recommended carrying volume: no more than 1.5 times the table top.
Refers to the force that the platform can bear when the work center of gravity is located in the center of the platform. If it is overloaded, it may make the platform move smoothly. The center of gravity of the objects carried on the surface of the water platform should not exceed the table surface. In order to ensure the movement accuracy and life of the platform, please check the allowable load of the platform when the platform is in use, and do not overload it.
Marking of stroke (moving amount)
For example: the stroke is ±12.7mm, as shown in the figure, the platform moves 12.7mm on one side and 12.7mm on the other side, and the total stroke is 25.4mm.
▶No recoil spring load design, detachable and interchangeable multiple actuators;
▶The platform can be completely locked with the fixing screws installed on the side;
▶When used with 20mm vernier micrometer, GON65-L-M provides ±5°rotational motion; GON65-U-M provides ±10°rotational motion;
▶Pure rotational movement around the virtual point;
▶There is a mark of 1° on the edge;
(1) Our angle stage can achieve unimpeded pure rotation movement around the rotation point within the rotation angle range. The stackable compact design allows the angle stage to be fixed to the upper surface of another angle stage, and the rotation points of the two angle stages overlap.
(2) The precise dovetail design, combined with the backlash-free screw and micrometer, enhances the angular positioning ability of the top plate, and provides accurate and repeatable positioning. Use the set screws on the side of the stage to lock the platform and lock the movement on the stage.
When used with a 20mm stroke vernier micrometer, the upper bracket of GON65-U-M has a rotational movement of ±10°, and the upper bracket of GON65-L-M has a rotational movement of ±5°. The hex screws on the side of the stage can be used to securely lock the movement on the stage.
(3) The angular displacement table has an array of threaded holes for installation, including 7 M6 and 4 M3 threaded holes. It is convenient to connect the opto-mechanical components to the stage. This series can be stacked through M6 countersunk mounting holes, or installed on a mini series breadboard.
Please note that we do not recommend removing the base plate from the angle stage. Because this may affect the precise adjustment of the stage, recalibration or readjustment is required.
Measurement error: Please allow a small measurement error.
- Additional Information